We present a novel bandwidth measurement scheme for multimode optical fibers. Amplified spontaneous emission (ASE) radiation was utilized for a source of intrinsically modulated light with a wide modulation bandwidth. In our measurement scheme, the continuouswave (CW) ASE light that passed through a multimode fiber (MMF) under test was analyzed by the fourthorder power with a highspeed photodetector and an electric spectrum analyzer. The modulation transfer function of the multimode fiber could be directly measured with the photoelectric spectra in the modulation frequency domain. The measurement result of our method was experimentally compared to that of the conventional measurement scheme based on the impulse response measurement. It has been found that our scheme provides a stable measurement means of MMF characterization that is suitable for the field testing due to the simplicity of the system.
I. INTRODUCTION
The multimode fiber (MMF) technology is widely used in shorthaul optical networks such as a localarea network (LAN) or a peripheral network
[1
,
2]
. In those optical communication systems, the typical length of the MMF links is a few hundreds of meters or shorter with usual bit rates of 1 to 10 Gbps. The MMF class has relative advantages of larger tolerances in fiber connection and the availability of the highspeed vertical cavity surface emitting lasers (VCSELs) that are better suited to the MMF
[1

3]
. Accurate evaluation of the MMF performance is indispensible in designing such a system. In contrast to the singlemode fibers (SMFs) where the groupvelocity dispersion (GVD) limits the maximum data rates, the chromatic dispersion plays little role in the MMF. It has a plurality of guiding modes that propagate at different group velocities with characteristic intermodal delays
[4

6]
. The link performance of the MMF can be properly described by the modulation bandwidth which is the frequencydomain measure of the intermodal dispersion. For this, several methods of the MMF bandwidth measurement have been developed and widely used in evaluation of the manufactured MMF cables as the standard methods
[5
,
6]
. It is worth stressing that each MMF cable even from a single preform needs to be evaluated in MMF bandwidth for reliable and predictable transmission performance
[2]
. By the nature of the MMF, a slight deviation of the refractive index profile can critically affect the modal dispersion in a sensitive manner.
It has been found that a number of factors can affect the actual bandwidth performance of the installed MMF link, not to mention the nonuniformity of the index profile along the fiber axis
[4
,
6
,
7]
. The light launch condition, operation wavelength, quality of splices or connectors, fiber bends and polarization states may considerably influence the actual bandwidth of the MMF link because they determine or change the mode power distribution and, consequently, influence the effective intermodal delays. This brings the needs for evaluating the bandwidth performance of the MMF links after the fiber has been installed in the fields. Particularly, field testing is highly demanded in upgrading the installed fiber networks for higher speeds when the information on the old MMF is insufficient. However, the current technologies used for the MMF bandwidth measurements do not provide desired simplicity and can rarely be used in field testing. Furthermore, as the MMFbased networks have evolved to Gigabit or 10Gigabit systems, the MMF links in the fields usually have very wide bandwidths above several GHz’s or higher
[1]
. It is technically more challenging than measuring long sections of manufactured MMF cables. A new bandwidth measurement method is demanded with a simpler system configuration and a wider effective measurement bandwidth, definitely above 10 GHz.
In this report, we present an MMF bandwidth measurement scheme that significantly simplifies the system configuration and the related maintenance issues. Our scheme uses a lowcost light source of amplified spontaneous emission (ASE) and a popular radiofrequency (RF) measurement instrument of the electric spectrum analyzer (ESA). Our scheme utilizes the ASE light that intrinsically exhibits random fluctuations of intensity which are determined by the optical spectrum. Due to the wide bandwidth along with the nearly uniform spectral characteristic of the ASE, the transmission characteristic of the MMF can be analyzed with ease by obtaining the photoelectric spectrum detected at the output of the MMF under test. Used in the MMF bandwidth measurement, the practically unlimited bandwidth of the ASE light minimizes the needs for careful calibrations of the light source and the concerns associated with the unwanted deviations. In this work, we successfully demonstrated the MMF bandwidth measurement and compared the result to that of the conventional scheme based on a shortpulse laser. It has been found that our scheme of the ASEbased MMF bandwidth measurement provides a simple and powerful test tool with a good measurement reliability, which will be useful in the field testing of MMF bandwidth as well as for a generic evaluation technique of the MMF cables.
II. MMF BANDWIDTH MEASUREMENT
The MMF bandwidth quantifies the modal dispersion that the lightwave signal experiences due to the intermodal delays of the modedependent group velocities
[5]
. Neglecting the chromatic dispersion effect that occurs for each mode, the transmitted signal is broadened in accordance with the modal characteristics. A signal input at an end of MMF is denoted by
f_{in}(t)
as a timedomain function of optical power. Then, the output signal obtained at the other end, denoted by
f_{out}(t)
, can be expressed by a sum of delayed Dirac delta functions of time
t, i.e
., a sum of
δ(t
𝜏
_{n}
), as
where
n
is the modal index of integer, 𝜏
_{n}
is the group delay, and
P_{n}
is the modal power for each mode of
n
, respectively. Here, ⊗ denotes an integral convolution. The
powerspread
model of Eq. (1) is an approximate way of understanding the MMF dispersion which is suitable for typical MMFs where the fieldsensitive phenomena are averaged out for a large number of modes. Eq. (1) can be rewritten in a simple form as
where
h(t)
is the characteristic function of the MMF, called as the impulse response in the linear system analysis. The impulse response is, then, a sum of the power transfer with modal delays as expressed by
with
P_{n}
’s of the normalized sum. The Fourier transform of Eq. (2) gives a simple algebraic formula as
where
F_{out}
,
F_{in}
and H are the Fourier conjugates of
f_{out}
,
f_{in}
and h in frequency
ω
, respectively. The transfer function of
H(ω )
is the Fourier transform of the impulse response,
h(t)
, and it determines the bandwidth of MMF. It is typically estimated by the frequency at which the amplitude of
H(ω )
decreases to 3 dB (50%) in respect to
H(0)
. This modulation bandwidth must be distinguished from an optical bandwidth. Notice that the description of Eq. (1) through Eq. (4) may include the effects of intermodal coupling and modedependent loss that the installed MMF may contain in the middle of the link. Those effects can be approximately characterized as exchange or decrease of the modal powers in the powerspread model of MMF.
The bandwidth of an MMF partially depends on the light launching condition that determines how the modes are excited. In Eq. (3), it can be represented by the mode power distribution of
P_{j}
. As a neutral launch condition, the classic method of the MMF bandwidth measurement utilizes the overfilled launch (OFL) in which the light field is incident with an excessively high numerical aperture (NA) with a low degree of spatial coherence
[5
,
6]
. Then, it can fully fill the angular and spatial components that excite virtually all the modes of the MMF with equal amounts of power as this can be described by a uniform value of
P_{n}
in Eq. (3). This OFL condition can be accomplished with ease for the case of incoherent light sources such as a light emitting diode (LED). Otherwise, especially with a laser source, a mode scrambler or mode mixer can be used, which strongly couples the guiding modes with intentionally applied local perturbations such as fiber bends. A near equilibrium of the mode power distribution can be established by this manner even with a highly coherent light source.
Based on the theory of modulation transfer, the MMF bandwidth can be analyzed in either time domain or frequency domain by utilizing a light source of a wide modulation bandwidth
[5]
. The most popular scheme of the MMF bandwidth measurement is to obtain the impulse response in time with impulselike short pulses. In an alternative approach, the transfer function can be directly obtained in frequency domain by using a light source modulated sinusoidally with a variable frequency. In both methods, the technical difficulty may arise in the source calibration because the modulation spectrum of the source is not uniform over the whole measurement frequency range and its bandwidth is finite. The modulation bandwidth of those light sources such as a gainswitched diode laser or an intensity modulatorequipped laser gives a few tens of GHz’s at best. Thus, the reliability of the measured result crucially depends on the stability of the modulated light source and the calibration accuracy, especially when the measurement range is wider than 10 GHz.
In this research on the MMF bandwidth measurement, we utilized an ASE light in continuouswave (CW) operation instead of using a modulated light source. The ASE light consists of continuously distributed spectral components of random phases which produce crossbeating products also known as the relative intensity noise
[8]
. The modulation spectrum after the downconversion process of a squarelaw detector can be found by the autocorrelation function of the optical spectrum. Hence, the modulation bandwidth of the ASE is as wide as the optical bandwidth of the spectrum. A very stable modulation characteristic can be obtained when the spectrum is determined by a passive element of an optical filter. Then, the modulation transfer function of an MMF can be obtained by comparing the modulation spectra measured at the different ports of the MMF under test.
The principle of our measurement scheme may need a theoretical justification. The intrinsic intensity modulation of the ASE fields has a different nature from the modulated lightwaves such as a pulsed laser or externally modulated light. It is not automatically accepted that the intensitymodulated contents of the ASE are temporally dispersed in the same way as those of the modulated lightwave signals. A suitable description of the ASE light fields that pass through an MMF is obtained from the
spectral intensity interferometer
[8]
. Applying this theory, the MMF can be understood as an interferometer of multiple light paths like a multiarm MachZehnder interferometer (MZI). Each path of a guiding mode or a coupled mode provides a characteristic delay.
Figure 1
shows the schematic diagrams that illustrate the MMF with an ASE source and a photodetector (a), the corresponding multiarm MZI (b), and the contents of the ASE light at the source part (c), respectively. Both of the MMF and the MZI give the multipath by which the light fields of the ASE are dispersed with time delays of 𝜏
_{n}
’s. The modulation spectrum of the ASE is an integration of the twocomponent beatings and makes a random fluctuation of intensity as explained by
Fig. 1(c)
. Our bandwidth measurement system measures the intermodal delays from the change of the intensity spectra induced by the multipath of the MMF.
Schematic diagrams that illustrate the MMF with an ASE source and a photodetector (a), the corresponding multiarm MZI (b), and the contents of the ASE light at the source part (c), respectively.
The MMF multipath inevitably brings an uncontrolled variation of the polarization states and path length fluctuations which may sensitively affect the output fields in conjunction with mode coupling or intermodal interference. The spectral intensity interferometry suggests that, regardless of the fieldsensitive process, the fourthorder power spectrum of the resultant field at the multipath output conveys only the intensityinvolved information
[8]
. This convenience is obtained by a narrow bandwidth of the ASE which must be less than 1/4 of the center frequency of the optical spectrum. The usual ASE light sources surely fulfill this narrowband condition. Therefore, the photoelectric power spectrum of a photodiode, which is the forth power of the ASE field, is independent of the fieldsensitive interference that occurs in the MMF.
Interferometric measurement of the intermodal delays or the modal dispersion of an MMF is not a completely new concept. It has been revealed that the optical frequencydomain reflectometry or the spectral interferometry can measure the modal characteristic in an interferometric configuration
[9

11]
. However, those secondorder interferometers sense the fieldsensitive interference so that they are naturally vulnerable to the phase and polarization variations of the modes. Despite some attractive features of the secondorder interferometry, the result is hardly translated to the MMF bandwidth without
a priori
knowledge on the MMF modes. Our ASEbased measurement method that utilizes the fourthorder interferometry is exclusively sensitive to the intensity contents that can accurately evaluate the bandwidth performance of the intensitymodulated communication link.
A more detailed description on our ASEbased measurement method is presented here to clarify the operation principle. Any lightwave of
E(t)
can be expressed with the center frequency of the spectrum
ω_{0}
as
where
A(t)
is the complex amplitude of the wave field. The ASE light has a broad spectrum centered at
ω_{0}
with a random phase at each frequency. The intensity, denoted by
I(t)
, is fluctuating as a consequence. The spectrum of the intensity is found by the Fourier transform of I(t). Denoting the Fourier transform with
F
{⋯}, the intensity spectrum at the source,
I_{0}(t)
, has its intensity spectrum as
where
Ĩ_{0}
and
Ã
are the Fourier conjugates of
I_{0}
and
A
, respectively. Eq. (6) suggests that the intensity spectrum is the spectral autocorrelation function of the amplitude spectrum. This ASE light goes through the multipath given by the MMF. From the result derived in Ref. 8, the fourthorder spectrum
S
^{(4)}
in modulation frequency domain is found under the narrowband condition as
when detected at the MMF output. Here,
n
and
m
are the mode indices, and the spatially averaged optical intensity of a mode of
n
is denoted by
𝑎_{n}
․
I_{0}(t)
with a positive real coefficient of an. The relative delay of 𝜏
_{nm}
is defined by the temporal difference of 𝜏
_{n}
𝜏
_{m}
between the two modes. It is assumed, in Eq. (7), that the intensity spectrum at the output is the same as
Ĩ_{0}
for each mode, which implies neglecting the GVD effect for the MMF.
From Eq. (3), the transfer function of the MMF is derived to be a sum of the transformed delta functions as
by the property of the Fourier transform with the mode index
n
. Its amplitude is, thus, found by
when
P_{n}
=
𝑎_{n}
is taken for simplicity. Therefore, the squared absolute transfer function of Eq. (9) is proportional to the fourthorder spectrum of
S
^{(4)}
given in Eq. (7). The amplitude of the transfer function is obtained easily from the experimental data as
in our ASEbased measurement method. The power of
Ĩ_{0}
can be obtained in a separate measurement performed with no MMF under test in place.
III. EXPERIMENT
In this investigation, we constructed an ASEbased MMF bandwidth measurement system with an ASE light source and an electric spectrum analyzer. The measurement result was compared to that of the conventional method based on a shortpulse laser.
Figure 2
shows the schematic diagrams of the filteredASE source (a), our ASEbased MMF bandwidth measurement system (b), and that of the pulsebased measurement system that we used for comparison. As shown in
Fig. 2(a)
, a semiconductor optical amplifier (SOA) produced ASE light and also reamplified the ASE backreflected by a fiberoptic reflector for a higher intensity. After an optical isolator, a bandpass filter (BPF) was installed to reduce the optical bandwidth to 20 nm, centered at 1,310 nm in wavelength. The intensity was adjusted with an optical attenuator. All the fiberoptic components in the source part used singlemode fiber for interconnection.
Schematic diagrams of the filteredASE source (a), our ASEbased MMF bandwidth measurement system (b), and that of the pulsebased measurement system that we used for comparison.
In our measurement system described in
Fig. 2(b)
, the ASE light fed by the source part went through a conditioning MMF for mode scrambling. A series of perturbations including bends and offset splices were applied to the conditioning MMF of a highNA stepindex fiber. The modescrambled light was launched to a fiber under test (FUT), a gradedindex MMF designed for optical communications, through a fiber connector. A photoreceiver connected to the FUT converted the received intensity signal to the photocurrent and fed it to an electric spectrum analyzer(E4440A, Agilent Technologies Inc.) that acquired the power spectrum of the photocurrent. The photoreceiver used in the system equipped a highspeed photodiode and an internal amplifier with a combined bandwidth of 11 GHz. For comparison, the conventional impulse response measurement system was constructed by using the same mode scrambler and photoreceiver. A gainswitched laser diode of distributed feedback (DFB) laser was used for the shortpulse source while a wideband sampling oscilloscope (DCA86100A, Agilent Technologies Inc.) acquired the received timedomain response. The center wavelength of the DFB laser was 1310.3 nm with a pulse repetition rate of 10 MHz. In both methods, the backtoback measurements were performed separately to obtain the system characteristic while no FUT was in place.
A 550m long MMF of core diameter 62.5 μm was used as the FUT. The impulse response measured with the system described in
Fig. 2(c)
showed the intuitive picture of the intermodal delays.
Figure 3
shows the acquired impulse response of the MMF (solid line), displayed with the system response measured in the backtoback condition (dotted line). The measured pulse width of the system response was 55 ps in full width at half maximum (FWHM) for the dotted line of
Fig. 3
. In the impulse response of the MMF, the clearly observed two peaks suggested that the modal delays were distinguished by two mode groups while each group was dispersed slightly. The Fourier transforms of the responses gave the transfer function of
H(ω )=F_{out}/F_{in}
as suggested by Eq. (4) where
F_{out}
and
F_{in}
are the Fourier transforms of the impulse response and backtoback system response, respectively.
Acquired impulse response of the MMF (solid line), displayed with the system response measured in the backto back condition (dotted line).
Our ASEbased measurement system described in
Fig. 2(b)
directly acquired the power spectrum with the spectrum analyzer. The same FUT was used while the optical power incident onto the FUT was adjusted to 0.5 mW.
Figure 4
shows the screencaptured image of the electric spectrum analyzer with three traces: the noise level acquired with no optical input (A), the spectrum measured while the FUT in place (B), and the spectrum without the FUT in the backtoback condition (C). The decrease of the spectrum C was caused by the limited bandwidth of the photoreceiver which affected commonly both of the photoelectric spectra of B and C. Note that the acquired spectral data were all on the power of the photocurrent or the fourthorder power of the light fields. The optical intensity spectrum could be obtained by taking the square root of the acquired electric power. The transfer function could be obtained in this way. Instead, for minimization of the noise effect, an alternative formula was used which is given as
where
G_{out}, G_{in}
and
G_{noise}
are the acquired power spectra of the FUT response, the backtoback response and the noise, respectively. Because the incoherent random noise of
G_{noise}
was added as an electric noise power to the spectra, the noise spectrum was subtracted before taking the square root in Eq. (11). This enhanced the measurement accuracy for a given power of the ASE.
Screencaptured image of the electric spectrum analyzer with three traces: the noise level acquired with no optical input (A), the spectrum measured while the FUT in place (B), and the spectrum without the FUT in the backtoback condition (C).
Figure 5
shows the normalized transfer functions measured by the two methods of our ASEbase measurement (solid line) and the impulse response measurement (dotted line) for the 550m MMF sample. Each was normalized so that the amplitude at frequency zero or the vicinity became a unity,
i.e., H
(0)=1. Those two transfer functions showed very similar features. The cosinelike fluctuations must have come from the presence of the two mode groups that formed the distinct peaks in the impulse response observed in
Fig. 3
. The interval of the fluctuation was ~5 GHz as seen in
Fig. 5
which well matched the separation of the impulse response peaks, ~0.2 ns, found in
Fig. 3
. The 3dB bandwidth was difficult to determine with a certain value because of the fluctuating feature of the transfer function but could be estimated to be ~3 GHz by the first point that crossed or nearly crossed the discrimination level of 3 dB (50%). The bandwidthlength product was estimated to be ~3 GHz by 0.55 km or ~1.65 GHz · km for the MMF sample.
Normalized transfer functions measured by the two methods of our ASEbase measurement (solid line) and the impulse response measurement (dotted line) for the 550m MMF sample.
The measurement repeatability was tested for the two methods. From the data obtained by five successive measurements performed in the same condition, a statistical deviation of the measured transfer functions was evaluated for each method. The interval between the consecutive measurements was ~60 seconds. A single backtoback measurement was performed to obtain
F_{in}
or
G_{in}
in each case, which was used for calculation of the transfer functions. In the impulse response measurement system, each timedomain impulse response was acquired with a waveform average number of
N
=16 while the backtoback response was acquired with
N
=256 for reduced noise. In our ASEbased measurement system, each spectrum trace was acquired with a sweep time of 1 s while the system characteristics including the noise spectrum were acquired with a longer acquisition time of 25 s. Note that the backtoback response of our ASEbased method was dominantly made by characteristics of the photoreceiver and the spectrum analyzer and does not need to be frequently calibrated.
Figure 6
shows the spectral standard deviation of the repeated measurements for the case of the ASEbased system (solid line) and that of the impulse response measurement system (dotted line). Spending similar acquisition times, our ASEbased measurement system exhibited significantly lower deviations especially for the highfrequency region above 10 GHz. The rapid increase of the error in the impulse response measurement could be explained by the limited bandwidth of the system response.
Figure 7
shows the spectrum of the system response,
F_{in}
(solid line) and its 1/2’th power,
(dotted line) for the backtoback system response measurement. The dotted curve in
Fig. 7
showed a similar characteristic to that of the standard deviation of the impulse response measurement shown with the dotted line of
Fig. 6
. This suggested that the error of the transfer function measurement is inversely proportional to the square root of the spectral amplitude. Hence, the limited modulation bandwidth of the pulsed source was understood to result in considerable inaccuracy in the transfer function measurement.
Spectral standard deviation of the repeated measurements for the case of the ASEbased system (solid line) and that of the impulse response measurement system (dotted line).
Spectrum of the system response, F_{in} (solid line) and its 1/2’th power, F_{in}^{1/2}(dotted line) for the backtoback system response measurement.
By using our ASEbased system, measurements of a short fiber section were also performed with a 100m MMF that originated from the same fiber spool.
Figure 8
shows the normalized transfer functions obtained by five successive measurements for the 100m MMF. Very small deviations were observed over the whole frequency range up to 15 GHz. The 3dB bandwidth was evaluated to be 15 GHz or slightly higher, which makes a bandwidthlength product of ~1.5 GHz · km for the 0.1km section of the MMF. This agreed with the measurement result of the 550m MMF, ~1.65 GHz · km. The difference of 10% was accounted for the intermodal coupling effect of the tightly spooled MMF in part
[4]
. The stable measurement result obtained by our method with the short MMF section demonstrated the feasibility of our measurement scheme in the field test where >10 GHz is needed for the measurement bandwidth.
Normalized transfer functions obtained by five successive measurements for the 100m MMF.
One more advantage of our ASEbased MMF characterization scheme is found in its wide wavelength adjustability. Due to the wide optical bandwidth of the ASE source spectrum, a wavelength range of ~50 nm can be covered by a single SOA light source, which can easily extended further with multiple SOAs or superluminescent diodes. The adjustability can be obtained with ease by changing the center wavelength of the optical filter used in the source part. We measured the wavelength dependence of the MMF bandwidth by utilizing four different BPFs with a nominal wavelength spacing of 20 nm.
Figure 9
shows the normalized transfer functions measured with the 550m MMF sample at different wavelengths of 1,272 nm, 1,288 nm, 1,308 nm and 1,326 nm. Here, the wavelength was evaluated by the mean wavelength of the filtered ASE spectrum. The experimental result showed that the transfer characteristic did not vary considerably for the lowfrequency band below 2 GHz but, in contrast, it exhibited noticeable wavelength dependence in the highfrequency range. This feature of our ASEbased measurement scheme must be useful when the wavelength dependence of the bandwidth matters in the applications.
Normalized transfer functions measured with the 550m MMF sample at different wavelengths of 1,272 nm, 1,288 nm, 1,308 nm and 1,326 nm, respectively.
IV. DISCUSSION
In this research, we showed that our MMF bandwidth measurement scheme has several advantages compared to the conventional measurement techniques. Our method utilizes a lowcost ASE light source without any modulation electronics, which greatly simplifies the measurement system and improves the measurement reliability. The measurement bandwidth can be easily extended to tens of GHz or higher owing to the wide bandwidth of the ASE light and the availability of the wideband electric spectrum analyzer that easily covers 40 GHz or wider. The detected spectral density of the fourthorder power spectra can be further improved by reducing the optical bandwidth for a given total optical power. Then, with a higher signaltonoise ratio, our system could measure the backreflected responses of FUTs to double the measurement accuracy in a reflectometric configuration
[12]
.
Our method has two unresolved issues that the present research has not revealed yet, and may need further investigations in separate studies. The first issue is related to the chromatic dispersion effect in the measurement result. In finding the transfer functions, the effect of GVD was neglected for simplicity as the conventional MMF bandwidth measurement also does. It could be justified by the fact that the modal dispersion is usually far larger than the GVD for typical MMFs. An MMF mostly has an intermodal delay definitely larger than 100 ps/km while the chromatic dispersion can be as large as 100 ps/nm/km at the short wavelength band of 850 nm
[1]
. At the 1,310nm band, it is surely negligible, below 5 ps/nm/km. But, in some cases, the measured MMF bandwidth can be considerably contaminated by the GVD effect that should be minimized or compensated
[1
,
3]
. At least, a rough estimation on the GVD effect should be provided for the ASE field that is obtained through an MMF.
The second issue of our ASEbased method is related to the absence of the phase information in our spectrum measurement. Our ASEbased measurement system detects the amplitude of the transfer function in losing the phase information that is essential for reconstructing the temporal contents of the impulse response via the inverse Fourier transform. In this context, our measurement method is unable to measure the intermodal delays but evaluates the bandwidth only, given in the frequency domain. This limitation may be a critical point of our measurement method because of the importance of the current laseroptimized MMF class for highspeed communications.
It has been found that the laserirradiated MMF exhibits considerably different performances in actual communication systems from what the OFL bandwidth predicts
[6]
. The launch condition of the laser light usually makes a restricted mode launch (RML) without a mode scrambling mechanism. The differential modal delay (DMD) has been introduced for measuring the modal delays by differentiating the launch condition. The most well established way of differentiation is to translate the launching beam spot across the endface of the MMF sample while the probe light is incident through the small beam onto the MMF
e.g
. by utilizing a scanned SMF launch. The obtained DMD data can be used to calculate the effective modal bandwidth (EMB or EMBc) with a set of the weightings that simulate certain types of the transmitters. Here, the last letter of ‘c’ in EMBc stands for ‘calculated’ when it is estimated from the DMD data. In the standard characterization method
[6]
, the minimum value of the EMBc (
i.e
. minEMBc) stands for the bandwidth performance for guaranteed performance. The value of minEMBc is determined with the ten standard sets of weighting functions. Those rather complicated and timeconsuming evaluation methods reflect the complexity of the MMF behavior with a nonspecific transmitter. For the DMD or EMBc measurements, the complete information of the modal delays is required for further processing of the data.
To cope with the EMBc standard, we suggest three possible ways of utilizing our ASEbased measurement method for the highspeed MMF links with a particular emphasis on the field testing. The simplest way is to use our ASEbased OFL measurement as a rough estimator for the modal delays in an RML condition. The OFL bandwidth, itself, can still give useful information to compare the MMF installed in the field with the manufacturer’s data. The careful analysis on the cosine fluctuations observed in the transfer function, if any, may give further information that is useful for estimating the actual link performance.
The second way is to introduce an optical mask for simulating a certain type of transmitters. This mask can be inserted between the ASEirradiating SMF and the FUT and forms an incident field similar to the field distribution of the laser transmitter. An electrically controlled spatial light modulator (SLM) can be used for such a selective mode excitation with a capability of rapidly changing the weighting functions for a direct EMB measurement
[13]
. This approach can be highly preferable method in the field test, especially when the characteristics of the transmitters to be used are known. The third possible way is to develop an ASEbased measurement system that can measure the phase contents that will allow us to obtain the DMD data through the inverse Fourier transforms. Adding a separate reference arm that delivers the reference light of ASE in an SMF can be considered as in an MZI configuration
[8
,
10
,
11]
. But this approach increases the system complexity and may make our ASEbased method less attractive for the field testing.
V. CONCLUSION
In this research on the MMF bandwidth measurement, we introduced a novel MMF measurement scheme based on an ASE light source as a selfmodulated light source. Through the experimental studies, we have found that our ASEbased method provides a wide measurement bandwidth with a small variation of the source characteristic when compared to the conventional impulse response measurement method. In our measurement system, the compactness of the source part which consists of passive fiberoptic components reduces the system complexity. In the field testing, the source part could be separated from the detection part being located at the other end of the FUT because of the nonsynchronous nature of the measurement. The wavelength adjustability was also found to be an attractive feature of our method for the evaluation of wavelengthdependent MMF bandwidths. Due to those advantageous features, our ASEbased MMF bandwidth measurement scheme can be a promising alternative to the conventional pulsebased techniques. It is worth noting that the 850nm band is the major transmission window for the dominant highspeed MMF links. Further investigations on the 850nm ASEbased measurement can enable fullband characterization of the MMF transmission capability.
Acknowledgements
This work has been supported in part by the Ministry of Education Science Technology (MEST) through the National Research Foundation of Korea (Grant No. 012R1A4A 102906) and by the Institute of Physics and Applied Physics at Yonsei University through BK21plus program
Molin D.
,
BigotAstruc M.
,
Kuyt G.
,
Melin G.
,
Sillard P.
2012
“Multimode fibers for costeffective highspeed, shortrange networks,”
European Conference and Exhibition on Optical Communication
2009
TIA Specification for 850nm Laser Optimized OM4 MMF, TIA492AAAD
Castro J. M.
,
Pimpinella R.
,
Kose B.
,
Lane B.
2012
“Investigation of the interaction of modal and chromatic dispersion in VCSELMMF channels,”
J. Lightwave Technol.
http://dx.doi.org/10.1109/JLT.2012.2203351
30
2532 
2541
Midwinter J. E.
1992
Optical Fibers for Transmission
Krieger Publishing Company
Malabar, USA
Miller C. M.
,
Derickson D.
1998
Fiber Optic Test and Measurement
Prentice Hall
Upper Saddle River, USA
“Intensity modulation and noise characterization of optical signals,”
246 
283
Ellis R.
2007
“The importance of minEMBc laser bandwidth measured multimode fiber for high performance premises networks, White Paper WP1150
Corning Inc
Corning, USA
http://www.corning.com/WorkArea/linkit.aspx?LinkIdentifier =id&ItemID=7881.
Shemirani M. B.
,
Mao W.
,
Panicker R. A.
,
Kahn J. M.
2009
“Principal modes in gradedindex multimode fiber in presence of spatial and polarizationmode coupling,”
J. Lightwave Technol.
http://dx.doi.org/10.1109/JLT.2008.2005066
27
1248 
1261
Moon S.
,
Noh H.
,
Kim D. Y.
2013
“Frequencydomain acquisition of fourthorder correlation by spectral intensity interferometry,”
Opt. Express
http://dx.doi.org/10.1364/OE.21.023206
21
23206 
23219
Ahn T.J.
,
Moon S.
,
Youk Y.
,
Jung Y.
,
Oh K.
,
Kim D. Y.
2005
“Mode analysis and modal delay measurement of a fewmode fiber by using optical frequency domain reflectometry,”
Journal of the Optical Society of Korea
http://dx.doi.org/10.3807/JOSK.2005.9.2.054
9
(2)
54 
58
Ahn T.J.
,
Moon S.
,
Kim S.
,
Oh K.
,
Kim D. Y.
,
Kobelke J.
,
Schuster K.
,
Kirchhof J.
2006
“Frequencydomain intermodal interferometer for the bandwidth measurement of a multimode fiber,”
Appl. Opt.
http://dx.doi.org/10.1364/AO.45.008238
45
8238 
8243
Lee J. Y.
,
Ahn T.J.
,
Moon S.
,
Youk Y. C.
,
Jung Y. M.
,
Oh K.
,
Kim D. Y.
2006
“Fourierdomain lowcoherence interferometry for differential mode delay analysis of an optical fiber,”
Opt. Lett.
http://dx.doi.org/10.1364/OL.31.002396
31
2396 
2398
Moon S.
,
Kim D. Y.
2009
“Reflectometric fiber dispersion measurement using a supercontinuum pulse source,”
IEEE Photon. Technol. Lett.
http://dx.doi.org/10.1109/LPT.2009.2025381
21
1262 
1264
Carpenter J.
,
Wilkinson T. D.
2012
“Characterization of multimode fiber by selective mode excitation,”
J. Lightwave Technol.
http://dx.doi.org/10.1109/JLT.2012.2189756
30
1386 
1392