A stack of gradientindex (GRIN) rod lenses cannot be used for coherent antiStokes Raman scattering(CARS) microendoscopy for insertion to internal organs through a surgical keyhole with minimal invasiveness. That’s because GRIN lens has large amount of inherent chromatic aberrations in spite of absolutely requiring a common focus for pump and Stokes beam with each frequency of ω
_{p}
and ω
_{S}
. For this endoscopic purpose, we need to develop a long slender probetype objective, namely probetype microscope objective (PMO). In this paper, we introduce the structure, the working principle, and the design techniques of PMO which is composed of a probetype lens module (PLM) and an adaptor lens module(ALM). PLM is first designed for a long slender type and ALM is successively designed by using several design parameters from PLM for eliminating optical discords between scanning unit and PLM. A combined module is optimized again to eliminate some coupling disparities between PLM and ALM for the best PMO. As a result, we can obtain a long slender PMO with perfectly diffractionlimited performance for pump beam of 817 nm and Stokes beam of 1064 nm.
I. INTRODUCTION
Obtaining molecular imaging in living organisms is applicable to the diagnosis of diseases such as cancer and cardiovascular disease
[1

3]
. This imaging technique can also be utilized in treating these kinds of disorders by testing the clinical effects of new medication. Non linear optical (NLO) microscopy is a candidate being actively researched in this field. It can obtain the resolution of several hundreds of nanometers. So to speak, NLO microscopy is actively utilized to image the cellular details of many organs and disease processes including cell trafficking, cell migration, and molecular interactions
[4

8]
. The terminology of NLO microscopy actually includes all the microscopies based on multiphoton interactions such as twophoton excitation fluorescence (TPEF) microscopy, second harmonic generation (SHG) microscopy, sum frequency generation(SFG) microscopy, and coherent antiStokes Raman scattering(CARS) microscopy. Among these kinds of microscopies,coherent antiStokes Raman scattering (CARS) microscopy is preferred to other microscopies because of labelfree strong vibrational signals, namely no use of fluorescent dyes
[9

11]
. However, in spite of this merit of CARS microscopy, multimodality based on CARS microscopy is preferable to CARS only
[12

13]
, because different NLO microscopies each have their distinct advantages in visualizing morphological details.
When we are going to construct CARS microscopy, the microscope is absolutely required to be inserted into a living body. We know well that conventional microscope objective is so bulky that it cannot be inserted into a living body. To achieve this purpose with minimal invasiveness,we need a long slender probetype objective, namely a probetype microscope objective (PMO). That is, a PMO which can be inserted to internal organs through a surgical keyhole.
As a candidate for PMO, a stack of cylindrical gradientindex(GRIN) rod lenses can be easily imagined and considered. Researchers often utilize GRIN rod lenses for microscope probes
[14

20]
. There are mainly two cases :in vivo confocal microendoscopy connected with a fiber bundle and in vivo twophoton (or multiphoton) microendoscopy coupled with a conventional microscope objective.However, GRIN rod lenses are difficult to use in CARS microendoscopy unlike confocal or multiphoton microendoscopy,because the CARS signal is generated only at a coincidence focal point of the pump and Stokes beam with frequencies of ω
_{p}
and ω
_{S}
, respectively
[12]
. GRIN rod lenses have large amount of inherent chromatic aberration which cannot bring a focus for both frequency of ω
_{p}
and ω
_{S}
[21

22]
.
Meanwhile, there might be another choice of PMO,which is equipped with up to four lasers (488/561/633/748 nm) by Olympus Inc. for multicolor fluorescence. However,this PMO is also difficult to apply for CARS microendoscopy,because CARS microscopy chiefly uses near infrared wavelengths, namely 817 nm for pump beam and 1064 nm for Stokes beam. In addition to this reason, the PMO is not easily accessible from Olympus Inc. because of some patent problems. In order to stand at the leading edge of world class development, it is essential that an optical system development like PMO goes with molecular imaging technology. In this paper, we introduce the design procedures and techniques of PMO, especially for CARS imaging.
II. DESIGN IDEAS AND DESIGN REQUIRMENTS
Coherent antiStokes Raman scattering (CARS) signal arises only at a focal point by two laser pulses with the frequencies of both ω
_{p}
and ω
_{S}
, called pump and Stokes beam. This means that the signal efficiency is strongly dependent on chromatic aberration between two frequencies.When the beat frequency of ω
_{p}
 ω
_{S}
is tuned to be resonant with a given molecule vibration (ω
_{v}
), a strong antiStokes signal is coherently driven at ω
_{as}
= 2ω
_{p}
?ω
_{S}
(
Fig. 1
A)
[23]
.A major improvement since the first attempt of CARS microscopy by Duncan and coworkers in 1982
[24]
was the development of the use of near infrared pump (800 nm) and Stokes beam (1200 nm) by Zumbusch and coworkers
[25]
. This is because near infrared frequencies instead of visible ones can more effectively reduce scattering and photo damage to the in vivo sample. And it is important to realize that the sensitivity of CARS detection is limited by nonresonant contributions such as the nonresonant CARS signal (
Fig. 1
B) and twophoton enhanced nonresonant signal (
Fig. 1
C). The use of near infrared frequencies can avoid the twophoton enhanced nonresonant contribution and lead to a significant improvement in sensitivity. Therefore,the frequencies (or wavelengths) of pump and Stokes beams are determined by the research of Conor L. Evans and coworkers
[12]
who used 817 nm for pump beam and
Resonant and nonresonant contributions for CARSsignal. (A) Resonant CARS. (B) Nonresonant CARS due tothe thirdorder susceptibility where the dotted lines indicatevirtual states. (C) Twophoton resonance of pump beamassociated with the excited electronic state.
1064 nm for Stokes beam for multimodality based on CARS microscopy.
As mentioned above, CARS signal is generated at a coincident focus by the thirdorder electronic interaction between two pulses and molecular resonant vibration. It is known that resonant CARS signal is proportional to the square of induced thirdorder polarization, namely P
^{(3)}
as shown in Eq. (1). Therefore, the signal generation efficiency is proportional to the second order power of pump beam intensity (I
_{p}
^{2}
) times the first order power of Stokes beam intensity (I
_{s}
)
[23]
. This leads to the critical condition for tight focusing without chromatic aberrations, so that PMO has to be perfectly corrected on monochromatic and chromatic aberrations.
where,
X
^{(3)}
denotes induced thirdorder susceptibility for the electronic interaction. E
_{p}
(ω
_{p}
) and E
_{s}
(ω
_{s}
) denote the electric fields for pump and Stokes beam. And E
_{p}
(ω'
_{p}
) denotes the second interaction between pump beam and molecular vibration.
Let’s see the schematic working structure of CARS microscope.
Fig. 2
shows a typical schematic diagram of an intravital laser scanning CARS microscope
[26

27]
. As shown in
Fig. 2
, PMO is composed of probe lens module (PLM)and adaptor lens module (ALM), in which PLM can be inserted to an internal organ through a surgical keyhole with minimal invasiveness. And intermediate image plane exists to mutually couple ALM to PLM. When collimated beams are incident on ALM from a scanning unit, ALM is expected to make a focal locus on an intermediate image plane. PLM is also expected to transfer this focal locus to the sample with tight focusing. If the focal points are tightly
A typical schematic diagram of an intravital laserscanning CARS microscope with probetype microscopeobjective (PMO) which is composed of adaptor lens module(ALM) and probe lens module (PLM) where ECARSTPEF and SFG denote epidectected CARS twophotonexcitation fluorescence and sum frequency generationsignal respectively. D.M. dichroic mirror.
A short type microscope objective where H and H’denotes first and second principle point and EFL denoteseffective focal length.
made on the sample plane, CARS signal arises at an each point and returned back to the detector after through PLM and ALM by turns. Summarizing the role of PLM and ALM, we can say that PLM and ALM correspond to a stack of GRIN rod lenses and a conventional microscope objective in twophoton fluorescence microscopy, respectively.
We have decided on the structure of probetype microscope objective (PMO) which consists of PLM and ALM without a detailed explanation. If PMO is expected to consist of only one module as in
Fig. 3
, there is no way for PMO to be a long slender type. A miniaturized microscope objective might have an effective focal length of about 2 or 3 mm.This causes the diameter of the clear aperture to be at least 2.8 mm for a numerical aperture of 0.7 and short length of probe. In order to be a long slender type, the use of an effective focal length of 20 mm brings the diameter of the clear aperture to be 28 mm. Therefore, PMO has to consist of PLM and ALM and ALM has to make a
A long slender type microscope objective which iscomposed of adaptor lens module (ALM) and probe lensmodule (PLM) where NAO denotes numerical aperture atobject side (or intermediate image side) and NA denotesnumerical aperture at image side (or sample side).
released focusing which causes PLM to be a miniaturized aperture, as shown in
Fig. 4
. The rest of the design requirements are determined by the Olympus PMO. It has the specifications such as outer diameter of 3.5 mm, field of view of 0.22 mm, numerical aperture of 0.7 at sample side, and length of PLM of 25 mm.
III. OPTICAL DESIGN OF PROBE LENS MODULE
In order to design a long slender type, we must first consider two key factors: intermediate image height and numerical aperture (NA) on the sample side. Refer
Fig. 5
,which has already been designed by our design group for a coherent antiStokes Raman scattering (CARS) imaging catheter system with fiber bundle. As shown in
Fig. 5
,almost the whole of the clear aperture is determined by NA. This system requires a 10.4 mm total track length from the fiber bundle end to the sample plane. For a longer total track length, the object plane has to be far away from the first lens surface. This will cause the diameter of clear aperture to be larger than 1.9 mm. Considering the outer diameter of 3.5 mm with the probe length of 25 mm for Olympus PMO (probetype microscope objective)
[19
,
26]
, it can be reasonable to choose the maximum clear aperture of 3.0 mm allowing for steel barrel thickness of 0.25 mm. Then, the total track length of 10.4 mm can be lengthened up to the intermediate image height (η) of 1.5 mm corresponding to the half of the maximum clear aperture of 3.0 mm. This also leads to NAO of 0.05 by the magnification relationship of PLM. NAO of 0.05 means the released focusing for ALM in
Fig. 4
, which leads to high structural stability in tolerance level for the whole PMO.
Since the length of PLM has to be 25 mm, an air gap of 15 mm is inserted between the intermediate image plane and the first lens surface in
Fig. 5
.
Fig. 5
is reoptimized to reach a goal of a long slender type without aberrations by CODE V
[28]
. In optimization, NAO (numerical aperture at object side) is kept to 0.05 and a merit function is composed of the square summation of Seidel 3
^{rd}
order aber
A miniaturized microscope objective for CARSimaging catheter with fiber bundle which the magnificationrelationship is given to 'M' = NAO/NA = η’/η where 'M'denotes the absolute value of magnification η denotes halfthe diameter of fiber bundle and η’ denotes half field of view(FOV) . Effective focal length of this system is also given tobe 2.975 mm.
The final form of PLM with the length of 25.5 mm andthe diameter of clear aperture of 2.94 mm where θ meansangle between horizontal dashed line and principal ray.Effective focal length of this system is given to 2.34 mm.
rations, 1
^{st}
order chromatic aberrations, and general ray aberrations as in our previous research.
Fig. 6
shows the final form of PLM with the length of 25.5 mm and the diameter of clear aperture of 2.94 mm. And the final performances of
Fig. 6
are shown in
Fig. 7
,
Fig. 8
, and
Fig. 9
. The radius of Airy disk, namely R = 0.61 λ/NA,is given as 0.71 ㎛ for pump beam of 817 nm and 0.93 ㎛ for Stokes beam of 1064 nm.
Fig. 7
shows the focus quality in terms of longitudinal spherical aberration (LSA)and astigmatic field curve (AFC). LSA represents focus errors on the sample plane arising from an onaxis point source on the object plane (or intermediate image plane).Refer to
Fig. 6
. And AFC also represents focus errors on the sample plane arising from an offaxis end point,namely 1.47 mm. As shown in
Fig. 7
, focus errors are completely corrected for both LSA and AFC. Especially, it is noted that the maximum focus deviations between pump and Stokes beam are 0.06 ㎛ for LSA and 0.29 ㎛ for AFC.
Fig. 8
shows spot diagrams for onaxis, 0.7 field,and 1.0 field (offaxis end) on sample plane. As shown in
Fig. 8
, it is confirmed that there are almost no position deviations between pump and Stokes beam. And since the radius of 100 % of encircled energy is given to be half of 1.67 ㎛, namely 0.84 ㎛, it can be said that PLM has
The focus quality charts which are longitudinalspherical aberration (LSA) and astigmatic field curve (AFC).In charts 2.5E4 mm denotes 0.00025 mm or 0.25 μm. T1 orS1 means the curved focus locus of tangential plane (yz) orsagittal plane (xz) for Stokes beam of 1064 nm and T2 or S2denotes for pump beam of 817 nm.
Spot diagrams. (A) Onaxis ( Strehl ratio of 0.999 andRMS wavefront of 0.006 waves for pump beam of 817 nm).(B) 0.7 field (Strehl ratio of 0.989 and RMS wavefront of0.017 waves for pump beam of 817 nm). (C) 1.0 field (Strehlratio of 0.983 and RMS wavefront of 0.021 waves for pumpbeam of 817 nm). Red and blue marks denote Stokes andpump beam respectively.
nearly diffractionlimited performance without chromatic aberrations considering the criterions of the Airy disk.
IV. OPTICAL DESIGN OF ADAPTER LENS MODULE
After finishing the design of the probe lens module(PLM) as a long slender type, we have to design the adaptor lens module (ALM) to link between the scanning unit and PLM. As shown in
Fig. 2
, the two input laser beams with frequency of ω
_{p}
and ωs are synchronously scanned on ALM and then form of focal points on intermediate image plane.Refer to
Fig. 4
. The intermediate image plane becomes the object plane at the same time for PLM. In the design of ALM, the most difficult problem is to find how to link between the scanning unit and PLM. Let’s consider
Fig. 9
which shows the geometrical relationship between ALM and PLM. From a fundamental Euclidean geometry of the triangle formed of between optical axis and principal ray
[29

31]
, we can write Eq. (2).
By a little algebraic calculation, Eq. (2) can be expressed as Eq. (3).
Because alternate angles are equal, s can be expressed as Eq. (4).
And the diameter of the aperture stop is given to Eq.(5) from the definition of fnumber.
If we consider three parameters given from the designed data of PLM such as θ = 3.740°, η = 1.471 mm, and NAO = 0.052, Eq. (3) and Eq. (5) give 22.503 mm as effective focal length (f
_{a}
) of ALM and Eq. (5) gives 2.356 mm to the diameter of aperture stop (D). An important thing is still left now. We have to decide the location of the aperture stop by from the principal ray. The principal ray passes through the center of the aperture stop and then is refracted by ALM keeping angle (θ) of 3.740° in the direction of the optical axis, as shown in
Fig. 9
. As a result, we have to keep the following 5 design parameters in the optimization process: fa of 22.503 mm, D of 2.356 mm, θ of 3.740°, η of 1.471 mm, and NAO of 0.052.
As in the design of PLM, ALM needs initial data for
The geometrical relationship between ALM andPLM.
optimization. See
Fig. 10
which is extracted from Japanese patents.
Fig. 10
is a conventional microscope objective which consists of the front focal length (f
_{f}
) of 1969.1 mm and the rear focal length (f
_{r}
) of 46.8 mm. Because the front focal length can be negligible compared to the rear focal length, the front part of the optical system is excluded before optimization. As in the optimization of PLM,we minimize a merit function which is composed of the square summation of Seidel 3
^{rd}
order aberrations, 1
^{st}
order chromatic aberrations, and general ray aberrations. As a result of optimization, the final form of ALM and spot diagrams are presented in
Fig. 11
and
Fig. 12
. And the radius of Airy disk, namely R = 0.61 λ/ NA, is given as 9.52 ㎛ for pump beam of 817 nm and 12.40 ㎛ for
An initial data for optimization which is extractedfrom Japan patents.
The final form of ALM with fa of 22.503 mm D of2.356 mm θ of 3.740° η of 1.471 mm and NAO of 0.052.
Spot diagrams. (A) Onaxis ( Strehl ratio of 1.000and RMS wavefront of 0.001 waves for pump beam of 817nm). (B) 0.7 field (Strehl ratio of 0.999 and RMS wavefront of0.005 waves for pump beam of 817 nm). (C) 1.0 field (Strehlratio of 1.000 and RMS wavefront of 0.001 waves for pumpbeam of 817 nm). Red and blue marks denote Stokes andpump beam respectively.
The final form of probetype microscope objective(PMO) which consists of ALM and PLM with field of view of0.22 mm and numerical aperture of 0.7 at sample side.
Spot diagrams. (A) Onaxis ( Strehl ratio of 0.999and RMS wavefront of 0.005 waves for pump beam of 817nm). (B) 0.7 field (Strehl ratio of 0.993 and RMS wavefront of0.014 waves for pump beam of 817 nm). (C) 1.0 field (Strehlratio of 0.995 and RMS wavefront of 0.012 waves for pumpbeam of 817 nm). Red and blue marks denote Stokes andpump beam respectively.
Stokes beam of 1064 nm. Therefore, it can be also said that ALM has perfectly diffractionlimited performance without chromatic aberrations considering the criterions of the Airy disk.
We have completed designs for both PLM and ALM.It’s time to combine these modules to complete the whole system, namely probetype microscope objective (PMO). After combining these modules, we have optimized again the whole system to eliminate some coupling disparities between PLM and ALM. As a result, we present the whole diagram of PMO in
Fig. 13
. For this PMO, spot diagrams are also presented in
Fig. 14
. Conclusively, it can be said that the final PMO has nearly diffractionlimited performance without chromatic aberrations for pump beam of 817 nm and Stokes beam of 1064 nm.
V. CONCLUSION
For coherent antiStokes Raman scattering (CARS) microendoscopy for insertion into internal organs through a surgical keyhole with minimal invasiveness, we have designed a long slender probetype objective, namely probetype microscope objective (PMO) instead of a stack of GRIN rod lenses.The structure of PMO had to be composed of probetype lens module (PLM) and adaptor lens module (ALM). For the design of PLM, the core issue was to design a long slender objective, which is controlled by two key factors:intermediate image height and NA at the sample side. And for the design of ALM, the core issue was to couple between the scanning unit and PLM, which is controlled by five design parameters: f
_{a}
, D, θ, η, and NAO (
Fig. 6
).After completing designs for both PLM and ALM, the whole system, namely PMO, was optimized again to eliminate some coupling disparities between PLM and ALM. In conclusion, the final PMO has shown nearly diffractionlimited performance without chromatic aberrations for pump beam of 817 nm and Stokes beam of 1064 nm.
Acknowledgements
This work has been supported by 2010 Hannam University Research Grant.
Jang S. J
,
Kang J. H
,
Kim K. I
,
Lee T. S
,
Lee Y. J
,
Lee K.C
,
Woo K. S
,
Chung W. S
,
Kwon H. C
,
Ryu C. J
,
Choi T. H
,
Choi C. W
,
Lim S. M
,
Cheon G. J
2010
Application of bioluminescence imaging to therapeutic intervention of herpes simplex virus type I  thymidine kinase/ganciclovir in glioma
Cancer Letters
297
84 
90
Waerzeggers Y
,
Monfared P
,
Viel T
,
Winkeler A
,
Jacobs A. H
2010
Mouse models in neurological disorders: applications of noninvasive imaging
Biochimica et Biophysica Acta?Molecular Basis of Disease
1802
819 
839
Pogue B. W
,
Samkoe K. S
,
GibbsStrauss S. L
,
Davis S.C
2010
Fluorescent molecular imaging and dosimetry tools in photodynamic therapy
Methods in Molecular Biology
635
207 
222
Olivier N
,
LuengoOroz M. A
,
Duloquin L
,
Faure E
,
Savy T
,
Veilleux I
,
Solinas X
,
Debarre D
,
Bourgine P
,
Santos A
,
Peyrieras N
,
Beaurepaire E
2010
Cell lineage reconstruction of early zebrafish embryos using labelfree nonlinear microscopy
Science
329
967 
971
Wallace S. J
,
Morrison J. L
,
Botting K. J
,
Kee T. W
2008
Secondharmonic generation and twophotonexcited autofluorescence microscopy of cardiomyocytes: quantification of cell volume and myosin filaments
Journal of Biomedical Optics
13
064018 
FioriniDebuisschert C
,
Berline I
,
Metge G
,
Charra F
,
Mihaly M
,
Allain C
,
Bordeau G
,
TeuladeFichou M. P
2009
Twophoton microscopy: from the optimisation of fluorescent DNA labels to local probe scanning second harmonic generation microscopy
Nonlinear Optics Quantum Optics
38
271 
280
AitBelkacem D
,
Gasecka A
,
Munhoz F
,
Brustlein S
,
Brasselet S
2010
Influence of birefringence on polarization resolved nonlinear microscopy and collagen SHG structural imaging
Opt. Express
18
14467 
14473
Sartenaer Y
,
Dreesen L
,
Humbert C
,
Volcke C
,
Tourillon G
,
Louette P
,
Thiry P. A
,
Peremans A
2007
Adsorption properties of decyl thiocyanate and decanethiol on platinum substrates studied by sumfrequency generation spectroscopy
Surface Science
601
1259 
1264
Yoo Y. S
,
Lee D. H
,
Cho H
2007
Differential twosignal picosecondpulse coherent antiStokes Raman scattering imaging microscopy by using a dualmode optical parametric oscillator
Opt. Lett.
32
3254 
3256
Lu F
,
Zheng W
,
Huang Z
2009
Coherent antiStokes Raman scattering microscopy using tightly focused radially polarized light
Opt. Lett.
34
1870 
1872
Hajek K. M
,
Littleton B
,
Turk D
,
McIntyre T. J
,
RubinszteinDunlop H
2010
A method for achieving superresolved widefield CARS microscopy
Opt. Express
18
19263 
19272
Evans C. L
,
Potma E. O
,
Puoris'haag M
,
Cote D
,
P.Lin C
,
Xie X. S
2005
Chemical imaging of tissue in vivo with videorate coherent antiStrokes Raman scattering microscopy
Proceedings of the National Academy of Sciences of the United States of America
102
16807 
16812
Lim R. S
,
Kratzer A
,
Barry N. P
,
MiyazakiAnzai S
,
Miyazaki M
,
Mantulin W. W
,
Levi M
,
Potma E. O
,
Tromberg B.J
2010
Multimodal CARS microscopy determination of the impact of diet on macrophage infiltration and lipid accumulation on plaque formation in ApoEdeficient mice
Journal of Lipid Research
51
1729 
1737
Gobel W
,
Kerr J. N. D
,
Nimmerjahn A
,
Helmchen F
2004
Miniaturized twophoton microscope based on a flexible coherent fiber bundle and a gradientindex lens objective
Opt. Lett.
29
2521 
2523
Levene M. J
,
Dombeck D. A
,
Kasischke K. A
,
Molloy R. P
,
Webb W. W
2004
In vivo multiphoton microscopy of deep brain tissue
Journal of Neurophysiology
91
1908 
1912
Jung J. C
,
Mehta A. D
,
Aksay E
,
Stepnoski R
,
Schnitzer M.J
2004
In vivo mammalian brain imaging using oneand twophoton fluorescence microendoscopy
Journal of Neurophysiology
92
3121 
3133
Kim P
,
Puoris’haag M
,
Cote D
,
Lin C. P
,
Yun S. H
2008
In vivo confocal and multiphoton microendoscopy
Journal of Biomedical Optics
13
010501 
Flusberg B. A
,
Cocker E. D
,
Piyawattanametha W
,
Jung J. C
,
Cheung E. L. M
,
Schnitzer M. J
2005
Fiberoptic fluorescence imaging
Nature Methods
2
941 
950
Wang H
,
Huff T. B
,
Fu Y
,
Jia K. Y
,
Cheng J. X
2007
Increasing the imaging depth of coherent antiStokes Raman scattering microscopy with a miniature microscope objective
Opt. Lett.
32
2212 
2214
Harzic R. L
,
Riemann I
,
Weinigel M
,
Konig K
,
Messerschmidt B
2009
Rigid and highnumericalaperture twophoton fluorescence endoscope
Appl. Opt.
48
3396 
3400
Rim C. S
2007
Design of an endoscope objective lens with a high numerical aperture and a minimallyinvasive outer diameter
J. Korean Phys. Soc.
51
52 
64
Chen X
,
George N
2008
Resolution analysis of a gradientindex rod and a gradientindex lens array
Appl. Opt.
47
6190 
6201
Cheng J. X
,
Volkmer A
,
Book L. D
,
Xie X. S
2001
Epidetected coherent antiStokes Raman scattering (ECARS)microscope with high spectral resolution and high sensitivity
Journal of Physical Chemistry B
105
1277 
1280
Duncan M. D
,
Reintjes J
,
Manuccia T. J
1982
Scanning coherent antiStokes Raman microscope
Opt. Lett.
7
350 
352
Zumbusch A
,
Holtom G. R
,
Xie X. S
1999
Threedimensional vibrational imaging by coherent antiStokes Raman scattering
Phys. Rev. Lett.
82
4142 
4145
Alencar H
,
Mahmood U
,
Kawano Y
,
Hirata T
,
Weissleder R
2005
Novel multiwavelength microscopic scanner for mouse imaging
Neoplasia
7
977 
983
Cheng J. X
2007
Coherent antiStokes Raman scattering microscopy
Applied Spectroscopy
61
197A 
208A
CODE V version 10.0
Optical Research Associates Inc.
http://www.opticalres.com
Smith W. J
,
Chapter 6.
2001
Modern Optical Engineering
MacGrawHill
NY USA
Chapter 6.
Lee J.U
,
Yu S.M
2009
Analytic design procedure of threemirror telescope corrected for spherical aberrationcoma astigmatism and Petzval field curvature
J. Opt.Soc. Korea
13
184 
192
Kweon GyeongIl
2010
Panoramic image composed of multiple rectilinear images generated from a single fisheye image
J. Opt. Soc. Korea
14
109 
120